Does anyone know what am I doing wrong? Message 5 of 8. My code just wants to unmask some bits in a register, read the register information through a query, and then set the front panel of the analyzer back to Local mode. Just one last question: This could affect your program if it contains time outs.
|Date Added:||16 December 2012|
|File Size:||63.86 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 3 of 5. I suggest you develop a sequence of buttons that you press to take the measurement, then code in that sequence and verify that is works correctly. Message 2 of How did you know 15 bytes is the minimum read? Since HpA is too old that I can not find it’s driver any more.
Additionally, the output shows 0 when it should show bits 4 and 13’s values added together from the first write.
Why are there so many coercion dots in your VI snippet? I discovered this value through trial and error.
GPIB-USB HPa – Discussion Forums – National Instruments
If you look lzbview the manual for the device you will find a place where it describes the default response to a read. Here is the working version.
Here is my code: Message 1 of So, I think hp a should follow the earliest protocol. Ill have to look for that. That is why I was using Auto 0 thing.
HP A interface with MATLAB – Discussion Forums – National Instruments
You can try this driver: Message 4 of 8. Am I doing something wrong?
And unfortunately the result is 0 which seems incorrect or meaningless. Message 7 of Message 4 of 5. Ah yes, the 15 bytes This information comes from the programming manual. Message 2 of 8. Does anyone know what am I doing wrong? Below is the part of the program.
Message 6 of I now know enough to know that I have no clue about anything at all. Message 7 of 8. It runs fine, however as you can see above, I have to hardcode in “15” in order to reach the end of the output.
Agilent Technologies / Keysight Technologies hp3562a Analyzer
When I run the program, the error 0 will happen. It outputs in the output string as I hoped!
Neither the driver or the assistant will gp3562a you the ability to save files. I have a question on implemintaion of automatic spectrum measurement with results saving into a text file using HPa spectrum analyzer.
I am tryinng to get the data from the analyzer now. My support system ensures that I don’t look totally incompetent.